|By PR Newswire||
|March 17, 2014 04:15 PM EDT||
PLEASANTON, Calif., March 17, 2014 /PRNewswire/ -- On March 4, 2014, Thoratec Corporation initiated a voluntary worldwide Medical Device Correction in order to update its labeling and training materials for the HeartMate II® LVAS Pocket System Controller (the "Pocket Controller"). The following information is provided as a reinforcement of the initial release.
Failure to completely connect the driveline during the process of exchanging Pocket System Controllers has resulted in serious injuries and deaths. Patients who received the Pocket Controller as a replacement for an older model (EPC System Controller) may be at a higher risk of experiencing difficulty in the controller exchange process, as there are differences between the two controller models, especially differences related to the connection of the driveline. As of February 4, 2014, the rate of serious injury or death during the process of controller exchange for patients who converted to the Pocket Controller from the EPC Controller was 1.5% (8/523), while the rate for patients who received the Pocket Controller at the time of implant was 0.1% (1/1619).
Patients who are using the HeartMate II LVAS Pocket Controller should immediately contact their doctor for retraining on use of the device and to receive updated Patient Handbook information if they have not been contacted by their doctor already. Pocket Controllers can be identified by the following model information that shows on the label on the controller: REF 105109.
Adverse reactions or quality problems experienced with the use of this product may be reported to the FDA's MedWatch Adverse Event Reporting program either online, by regular mail or by fax.
- Online: http://www.fda.gov/medwatch/report.htm
- Regular Mail: use postage-paid FDA form 3500 available at: http://www.fda.gov/MedWatch/getforms.htm. Mail to MedWatch 5600 Fishers Lane, Rockville, MD 20852-9787
- Fax: 1-800-FDA-0178
Web site: http://www.thoratec.com
SOURCE Thoratec Corporation
Jan. 18, 2017 02:30 AM EST Reads: 1,810
Jan. 18, 2017 02:15 AM EST Reads: 3,203
Jan. 18, 2017 01:30 AM EST Reads: 4,199
Jan. 18, 2017 01:15 AM EST Reads: 4,872
Jan. 18, 2017 01:00 AM EST Reads: 2,012
Jan. 18, 2017 01:00 AM EST Reads: 6,045
Jan. 18, 2017 12:45 AM EST Reads: 6,251
Jan. 18, 2017 12:45 AM EST Reads: 5,949
Jan. 18, 2017 12:30 AM EST Reads: 2,784
Jan. 18, 2017 12:00 AM EST Reads: 7,739
Jan. 18, 2017 12:00 AM EST Reads: 2,258
Jan. 17, 2017 11:30 PM EST Reads: 4,353
Jan. 17, 2017 11:15 PM EST Reads: 4,939
Jan. 17, 2017 11:00 PM EST Reads: 605
The many IoT deployments around the world are busy integrating smart devices and sensors into their enterprise IT infrastructures. Yet all of this technology – and there are an amazing number of choices – is of no use without the software to gather, communicate, and analyze the new data flows. Without software, there is no IT. In this power panel at @ThingsExpo, moderated by Conference Chair Roger Strukhoff, Dave McCarthy, Director of Products at Bsquare Corporation; Alan Williamson, Principal ...
Jan. 17, 2017 10:30 PM EST Reads: 2,369